Year : 2021 | Volume
| Issue : 4 | Page : 419-423
A comparative study of the efficacy of Glidescope versus Macintosh direct laryngoscopy for intubation in pediatric patients undergoing cardiac surgery
Ibrahim Zabani, Mohammed AlHarbi, Abdulkarim AlHassoun, Shafat Iqbal, Dareen Al Amoudi, Sultan AlOtaibi, Hasan Saad
Department of Anesthesiology, King Faisal Specialist Hospital and Research Center, Jeddah, Saudi Arabia
Prince Saud Al Faisal, Ar Rawdah, Jeddah - 23433
Source of Support: None, Conflict of Interest: None
|Date of Submission||22-Jun-2021|
|Date of Decision||23-Jun-2021|
|Date of Acceptance||24-Jun-2021|
|Date of Web Publication||02-Sep-2021|
Background: The Glidescope is a novel, portable, reusable video laryngoscope that has provided superior laryngeal visualization to facilitate tracheal intubation, especially in the management of difficult airways. In this study, we aimed to compare the efficacy of the Glidescope (video-laryngoscope) against the Macintosh direct laryngoscope. Methods: Fifty patients were randomly selected via simple randomization using computer-generated random numbers, and sorted into two groups of 25 patients: the Glidescope group and the Macintosh group. We included pediatric patients undergoing cardiac surgery for the repair of congenital heart disease. Those with suspected difficult intubation, preterm babies with low body weight, and patients at risk of aspiration were all excluded. Results: Patients' baseline demographic and clinical characteristics were found to be comparable in the two groups. The mean intubation time was 24.1 ± 13.6 s in the Glidescope group, as compared to 18.1 ± 5.9 s in the Macintosh group. Blade insertion was easy in 92% and 96% of the patients in the Glidescope and Macintosh groups, respectively. Tracheal intubation was considered easy in 84% of the Glidescope group, compared to 92% of the Macintosh group. There was a statistically significant correlation between the ease of tracheal intubation and the used intubation method (rho = –0.35; P = 0.014). Conclusion: Our findings indicate that the Glidescope can be used as an efficient modality for obtaining successful intubations with no complications. Ease of tracheal intubation was the only outcome that was found to be affected by the used modality. Further investigations with proper sample sizes are needed.
Keywords: Airway, general anesthesia, intubation, laryngoscopy, pediatric
|How to cite this article:|
Zabani I, AlHarbi M, AlHassoun A, Iqbal S, Al Amoudi D, AlOtaibi S, Saad H. A comparative study of the efficacy of Glidescope versus Macintosh direct laryngoscopy for intubation in pediatric patients undergoing cardiac surgery. Saudi J Anaesth 2021;15:419-23
|How to cite this URL:|
Zabani I, AlHarbi M, AlHassoun A, Iqbal S, Al Amoudi D, AlOtaibi S, Saad H. A comparative study of the efficacy of Glidescope versus Macintosh direct laryngoscopy for intubation in pediatric patients undergoing cardiac surgery. Saudi J Anaesth [serial online] 2021 [cited 2022 Aug 20];15:419-23. Available from: https://www.saudija.org/text.asp?2021/15/4/419/325297
| Introduction|| |
Successful tracheal intubation is a very necessary skill for the management of critically ill patients. The glidescope has been developed to facilitate the tracheal intubation process with ease in all scenarios. Previous studies reported superiority of the glidescope in comparison to the standard direct laryngoscope. Furthermore, cardiac surgeries are usually associated with an increased risk of developing complications that can worsen the prognosis of the relevant patient and increase their risk of morbidity and mortality., Therefore, reduced hospital stay and intensive care unit (ICU) admissions, in addition to early extubation and mobilization are encouraged in these patients to reduce the risk of the development of potential complications and to alleviate the economic burden., Maintaining adequate and competent airways during intubation is a priority for anesthesiologists. A previous review by Cook et al. indicated that adverse events related to the management of airways during anesthesia for patients with cardiac arrests are the main cause for half of these cardiac events. Although most tracheal intubations are easily performed with no impairment of the airways, it has been reported that up to 8.5% of anesthesiologists may find it difficult to maintain proper airway management during intubation. This can lead to the development of serious complications such as mortality due to hypoxemic brain damage. Other complications can also include laryngospasm, bronchospasm and dental damage as a result of fierce and repetitive intubation. Therefore, it was essential to address this experience of performing anesthesiologists and to enhance the quality of the equipment that is used for the delivery of the intubation process.
The Macintosh laryngoscope is commonly used for performing tracheal intubation and is often associated with difficult intubation in 1-4% of cases. As a result, the Glidescope video laryngoscope (GVL, Glidescope®, Saturn Biomedical Systems Inc.) has been developed for use in cases of difficult intubation processes., Previous studies have shown its superiority over the Macintosh laryngoscope technique.,,, However, the comparison between the two modalities in terms of intubation time is controversial. Previous studies have shown that the GVL can significantly increase the period of intubation.,,, However, other investigations have reported that the intubation time difference between the two modalities is not significant,, and some even reported that GVL may significantly reduce the intubation period., GVL is also not always affordable and cannot be obtained in the required sizes and shapes to suit all patients. Few studies have investigated the differences between the Macintosh laryngoscope and GVL in pediatric cardiac surgeries. We aim to compare the efficacy of the Glidescope versus Macintosh direct laryngoscopy for intubation in pediatric patients undergoing cardiac surgery.
| Materials and Methods|| |
Study design and population
This is a randomized control study that included 25 patients in each group to detect a difference of at least 12 s in intubation between the two groups with a significant level of 0.05, and a power of 80%. Data collection was conducted at King Faisal Specialist Hospital and Research Center in Jeddah, Saudi Arabia. Fifty patients were randomly selected via simple randomization using computer-generated random numbers into two groups – the Glidescope group and the Macintosh group – with 25 patients in each group. The study was concluded upon the completion of the required sample size. We included pediatric patients undergoing cardiac surgery for the repair of congenital heart disease. The patient's body weight had to be within the range of 3 to 30 kg. Those with suspected difficult intubation, preterm babies with low body weight, and patients at risk of aspiration were all excluded. The study was reported using the CONSORT statement for randomized control trials.
The anesthetic techniques were standardized; patients were premedicated via midazolam 0.03 mg/kg IV. Induction of anesthesia was done via Ketamine 1.5- 2.5 mg/kg IV, Fentany l 5-10 μg/kg. Endotracheal intubation was performed using Rocuronium as a muscle relaxant with a dose of 0.5 mg/kg IV. Intubation was conducted by one senior anesthesiologist, to exclude variation in intubation skills, and the choice of the intubation device was dependent on the randomization program. The second operator was responsible for monitoring the time taken for intubation in seconds, the incidence of desaturation, incidence of airway injury, the ease of intubation, and recording the data in the excel spreadsheet alongside the patient medical number and name.
Data was collected, coded, tabulated, and analyzed using the SPSS 26.0 statistical package. Continuous variables were represented as means and standard deviation and nominal variables were presented as counts and percentages. The skewness and Kurtosis tests were used for testing the normal distribution of continuous variables. To compare patients based on the intubation method used, the Chi-square test (or Fisher's exact test, when appropriate) was used for categorical data while an independent t test was used for normally distributed continuous variables, and the Mann–Whitney H test was used for continuous variables not normally distributed. Moreover, the Spearman rank correlation coefficient (rho) was used to determine the relationship between the endotracheal intubation method and patient outcomes. Statistical significance was considered when the P was < 0.05.
The study was approved by King Faisal Specialist Hospital and Research Center Institutional Review Board (IRB) with approval number RC-J/483/40. An informed written consent was obtained from the guardian of the patients who were enrolled in the study to include their data in the study without revealing any personal information. All methods were performed in accordance with the relevant guidelines and regulations Declaration of Helsinki 1975. In addition, the clinical trial was registered in the UMIN Clinical Trials Registry (UMIN-CTR) with registration number UMIN000044474.
| Results|| |
Patients' baseline demographic and clinical characteristics were found to be comparable in two groups, with no statistically significant differences [Table 1]. The mean age in the Glidescope group was 29.3 ± 37.0 months, as compared to 25.2 ± 30.6 months in the Macintosh group. Similarly, the mean BMI in the Glidescope group was 14.6 ± 2.3 Kg/m2, as compared to 13.9 ± 2.5 Kg/m2 in the Macintosh group. All patients included in the study had a Class II Mallampati classification and most (Glidescope group = 88%; Macintosh group = 72%) did not have any anatomical abnormalities.
All measured outcomes are summarized in [Table 2]. The mean intubation time was 24.1 ± 13.6 s in the Glidescope group, as compared to 18.1 ± 5.9 s in the Macintosh group. Blade insertion was easy in 92% and 96% of the patients in the Glidescope and Macintosh groups, respectively. For ease of tracheal intubation, it was considered easy in 84% of the Glidescope group, compared to 92% of the Macintosh group. As for the quality of the view, it was excellent in most of the Glidescope group (84%), similar to the Macintosh group (88%). The Cormack–Lehane grade during laryngoscopy was I/II in 19/6 patients in the Glidescope group and was I/II in 16/9 patients in the Macintosh group. Intubation was attempted only once in the majority of the Glidescope group (96%) and most of the Macintosh group (88%). No complications were recorded in either group.
The correlation analysis showed a statistically significant correlation between the ease of tracheal intubation and the intubation method used (rho = –0.35; P = 0.014). However, the correlation between other outcomes and the used intubation method did not reach statistical significance [Table 3].
|Table 3: Correlation between measured outcomes and different intubation methods|
Click here to view
| Discussion|| |
In the present study, we evaluated the differences between Glidescope and Macintosh direct laryngoscopy for intubation in pediatric patients undergoing cardiac surgery, in terms of efficacy and clinical outcomes. Our results indicate that there are no apparent differences between the two groups in terms of complications, the number of intubation attempts, and ease of blade insertion. A previous study by Jafra et al. reported that the Glidescope group showed better laryngeal views and intubation difficulty scores when compared to the Macintosh laryngoscopy group in an adult population undergoing elective surgeries. Another randomized trial by Choi et al. also reported that in the Glidescope group, the percentage of glottis opening was significantly increased when compared to the Macintosh laryngoscopy group. Moreover, the Glidescope group had a significantly lower visual analog score and reduced intubation time than the other group, which indicated the superiority of the modality. Similar findings were also reported by a randomized simulation trial by Kim et al.
In the present study, no complications were noticed in the two groups. However, some cases have been previously published that report a possible incidence of laryngeal trauma, potential pharyngeal injury and infections.,, Previous studies have shown that the rate of successful intubation was similar between the two groups, ranging between 95.5 and 100% for both modalities.,,, First attempts at successful intubation processes were also comparable between the two groups. Jafra et al. reported that the rate of first attempts was significantly higher in the Glidescope group. In contrast, Choi et al. reported that similar rates of first successful intubation processes were found across the two groups. Ibinson et al. also indicated that Glidescope was significantly more efficient at obtaining successful intubation rates from first attempts, as compared to the direct laryngoscopy group.
The results of the correlation analysis shown that ease of tracheal intubation may be the only factor that can significantly affect the success of intubation via either of the two modalities. A previous investigation by Ayoub et al. reported that more successful intubations could be obtained when using the Glidescope over the Macintosh laryngoscopy when both modalities were performed by inexperienced medical students. This indicates that the Glidescope may be easier to use to perform successful intubation processes. Previous reports have also shown that the Cormack–Lehane grading was significantly enhanced with the Glidescope group than the conventional group,,,,,, which is consistent with our findings.
We also found that the time of intubation was longer in the Glidescope group than the Macintosh direct laryngoscopy group. This finding is consistent with the findings of previous investigations.,,, However, other reports have shown that the period may be similar between the two modalities,, and further reports have even reported that using the Glidscope can significantly reduce the time needed for performing successful intubation., The difference in the findings of these various reports may be attributed to many factors. First, the difference of experience between the anesthesiologists that used both modalities in the different studies could be a potential factor, which indicates the need to recruit anesthesiologists that are adequately experienced in order to ensure a better and sufficient judgment. Second, the difference in the camera scope may also constitute a reason for the differences. Third, observing the monitor is essential to viewing the vocal cords when using the Glidescope, whereas it can be easily done via simple visualization when using the Macintosh laryngoscopy. The structure of the Macintosh blade is also different, which can raise some issues when performing intubation. These factors should be considered by future investigations for a better validation of the evidence and better judgment of the differences in efficacy between the two modalities.
Although no significant differences in the baseline characteristics were observed between the two groups, our study was limited by the small sample size of both groups, which prevented us from adequately validating our reported findings. Therefore, future investigations are encouraged to enroll adequate populations for better validation and estimation of the potentially correlated factors for the efficacy of both modalities. Moreover, we recommend that the previously discussed factors, including the experience of the performing anesthesiologists of both modalities, should also be considered for a better judgment on the efficacy of either modality.
| Conclusion|| |
Our findings indicate that the Glidescope can be used as an efficient modality for obtaining successful intubations with no complications. However, we found that the intubation time was longer in the Glidescope group than the Macintosh laryngoscopy group. Ease of tracheal intubation was the only outcome that was found to be affected by the used modality. Further investigations with proper sample sizes are needed.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Rabiner JE, Auerbach M, Avner JR, Daswani D, Khine H. Comparison of GlideScope videolaryngoscopy to direct laryngoscopy for intubation of a pediatric simulator by novice physicians. Emerg Med Int 2013;2013:407547. doi: 10.1155/2013/407547.
Zhang X, Zhao X, Wang Y. Dexmedetomidine: A review of applications for cardiac surgery during perioperative period. J Anesth 2015;29:102-11.
Doenst T, Bargenda S, Kirov H, Moschovas A, Tkebuchava S, Safarov R, et al
. Cardiac surgery 2018 reviewed. Clin Res Cardiol 2019;108:974-89.
Cheng DC. Fast track cardiac surgery pathways: Early extubation, process of care, and cost containment. Anesthesiology 1998;88:1429-33.
Mittnacht AJ, Hollinger I. Fast-tracking in pediatric cardiac surgery--the current standing. Ann Card Anaesth 2010;13:92-101.
] [Full text]
Cook TM, Woodall N, Frerk C. Major complications of airway management in the UK: Results of the Fourth National Audit Project of the Royal College of Anaesthetists and the Difficult Airway Society. Part 1: Anaesthesia. Br J Anaesth 2011;106:617-31.
Janssens M, Hartstein G. Management of difficult intubation. Eur J Anaesthesiol 2001;18:3-12.
Benumof JL. Management of the difficult adult airway. With special emphasis on awake tracheal intubation. Anesthesiology 1991;75:1087-110.
Sakles JC, Rodgers R, Keim SM. Optical and video laryngoscopes for emergency airway management. Intern Emerg Med 2008;3:139-43.
Cooper RM, Pacey JA, Bishop MJ, McCluskey SA. Early clinical experience with a new videolaryngoscope (GlideScope) in 728 patients. Can J Anaesth 2005;52:191-8.
Serocki G, Bein B, Scholz J, Dörges V. Management of the predicted difficult airway: A comparison of conventional blade laryngoscopy with video-assisted blade laryngoscopy and the GlideScope. Eur J Anaesthesiol 2010;27:24-30.
Lim TJ, Lim Y, Liu EH. Evaluation of ease of intubation with the GlideScope or Macintosh laryngoscope by anaesthetists in simulated easy and difficult laryngoscopy. Anaesthesia 2005;60:180-3.
Narang AT, Oldeg PF, Medzon R, Mahmood AR, Spector JA, Robinett DA. Comparison of intubation success of video laryngoscopy versus direct laryngoscopy in the difficult airway using high-fidelity simulation. Simul Healthc 2009;4:160-5.
Kim HJ, Chung SP, Park IC, Cho J, Lee HS, Park YS. Comparison of the GlideScope video laryngoscope and Macintosh laryngoscope in simulated tracheal intubation scenarios. Emerg Med J 2008;25:279-82.
Powell L, Andrzejowski J, Taylor R, Turnbull D. Comparison of the performance of four laryngoscopes in a high-fidelity simulator using normal and difficult airway. Br J Anaesth 2009;103:755-60.
Hirabayashi Y, Otsuka Y, Seo N. GlideScope videolaryngoscope reduces the incidence of erroneous esophageal intubation by novice laryngoscopists. J Anesth 2010;24:303-5.
Wayne MA, McDonnell M. Comparison of traditional versus video laryngoscopy in out-of-hospital tracheal intubation. Prehosp Emerg Care 2010;14:278-82.
Ayoub CM, Kanazi GE, Al Alami A, Rameh C, El-Khatib MF. Tracheal intubation following training with the GlideScope compared to direct laryngoscopy. Anaesthesia 2010;65:674-8.
Song JG, Kwon MA. Use of the Airtraq (R) optical laryngoscope as a rescue airway device following failed awake flexible fiberoptic nasotracheal intubation of a patient with severe microsomia: A case report. Korean J Anesthesiol 2008;55:353-7.
Jafra A, Gombar S, Kapoor D, Sandhu HS, Kumari K. A prospective randomized controlled study to evaluate and compare GlideScope with Macintosh laryngoscope for ease of endotracheal intubation in adult patients undergoing elective surgery under general anesthesia. Saudi J Anaesth 2018;12:272-8.
Choi GS, Lee EH, Lim CS, Yoon SH. A comparative study on the usefulness of the Glidescope or Macintosh laryngoscope when intubating normal airways. Korean J Anesthesiol 2011;60:339-43.
Cooper RM. Complications associated with the use of the GlideScope videolaryngoscope. Can J Anaesth 2007;54:54-7.
Xue FS, Zhang GH, Liu J, Li XY, Yang QY, Xu YC, et al
. The clinical assessment of Glidescope in orotracheal intubation under general anesthesia. Minerva Anestesiol 2007;73:451-7.
Aziz MF, Healy D, Kheterpal S, Fu RF, Dillman D, Brambrink AM. Routine clinical practice effectiveness of the Glidescope in difficult airway management: An analysis of 2,004 Glidescope intubations, complications, and failures from two institutions. Anesthesiology 2011;114:34-41.
Sun DA, Warriner CB, Parsons DG, Klein R, Umedaly HS, Moult M. The GlideScope video laryngoscope: Randomized clinical trial in 200 patients. Br J Anaesth 2005;94:381-4.
Choi HJ, Kim YM, Oh YM, Kang HG, Yim HW, Jeong SH, et al
. GlideScope video laryngoscopy versus direct laryngoscopy in the emergency department: A propensity score-matched analysis. BMJ Open 2015;5:e007884.
Ibinson JW, Ezaru CS, Cormican DS, Mangione MP. GlideScope use improves intubation success rates: An observational study using propensity score matching. BMC Anesthesiol 2014;14:101. doi: 10.1186/1471-2253-14-101.
Malik MA, Subramaniam R, Maharaj CH, Harte BH, Laffey JG. Randomized controlled trial of the Pentax AWS, Glidescope, and Macintosh laryngoscopes in predicted difficult intubation. Br J Anaesth 2009;103:761-8.
Malik MA, Maharaj CH, Harte BH, Laffey JG. Comparison of Macintosh, Truview EVO2, Glidescope, and Airwayscope laryngoscope use in patients with cervical spine immobilization. Br J Anaesth 2008;101:723-30.
[Table 1], [Table 2], [Table 3]